Switching the mode of GSM scales FILIP
GSM scales FILIP can work
in two modes, and can be switched off permanently by a magnet:
As simple scales in the mode DISP, which does not switch on GSM module and the weight and battery condition can be read only on red LED display. In this mode, regular measurements are carried out and stored in the scale’s memory according to set intervals, which can be adjusted in the GSM mode via SMS, e.g. "timer=60,0" – measurements will be carried out once an hour. If measuring is set as once per 24 hours, the capacity of scales memory can be used for records for more than 1,5 year. If you switch the scales into the mode GSM with activated chip SIM, and you will turn on WEB in the scales, the entire history will be downloaded on WEB server, and will have access to it. Switching from this mode to the mode GSM can be carried out by putting the magnet to the scales on the period longer than 10 seconds in the time, when the scales is active. On the display, there will be the text "FSLP" and the scales will turn off. By repeated putting the magnet you can wake up the scales, and it will work in the GSM mode.
In the mode GSM, the entire functionality of the scales is supported. In the case that chip SIM in the scales will not be activated, the scales will display on the display the text "noOP" – mobile operator is not accessible. If the chip SIM is active in the scales, and on the display, there is the text "noSG" or "noOP", maybe the signal of mobile operator is weak. You have to connect to stronger antenna or to switch the scales into the mode DISP. The scales will carry out regular measurements even if the mobile operator is not available, and immediately, after it will be available, recorded data will be downloaded from the scales to the WEB server, and you will have access to these data. Switching from this mode into the mode DISP can be carried out by putting the magnet to the scales for the period longer than 10 seconds in the time, when your scales is active. On the display, there will be the text "FSLP" and the scales will turn off. By repeated putting the magnet, you will wake up the scales and it will work in the mode DISP. On the display, there will be the text "dISP" for informing, which mode the scales operate in.
Permanent switch-off of the scales can be done by putting the magnet to the scales for the period longer than 10 seconds, when the scales are active. On the display, there will be the text "FSLP" and the scales will turn off permanently. The switch-on can be carried out by repeated putting the magnet next to the scales.
Switching the modes of the scales can be done also by means of SMS, if the scales is in the mode GSM. A command "reset=0" switches the scales from the mode GSM to the mode DISP.